- Research article
- Open Access
- Open Peer Review
A comparison of the temporary placement of 3 different self-expanding stents for the treatment of refractory benign esophageal strictures: a prospective multicentre study
© Canena et al.; licensee BioMed Central Ltd. 2012
- Received: 27 January 2012
- Accepted: 24 May 2012
- Published: 12 June 2012
Refractory benign esophageal strictures (RBESs) have been treated with the temporary placement of different self-expanding stents with conflicting results. We compared the clinical effectiveness of 3 types of stents: self-expanding plastic stents (SEPSs), biodegradable stents, and fully covered self-expanding metal stents (FCSEMSs), for the treatment of RBES.
This study prospectively evaluated 3 groups of 30 consecutive patients with RBESs who underwent temporary placement of either SEPSs (12 weeks, n = 10), biodegradable stents (n = 10) or FCSEMSs (12 weeks, n = 10). Data were collected to analyze the technical success and clinical outcome of the stents as evaluated by recurrent dysphagia, complications and reinterventions.
Stent implantation was technically successful in all patients. Migration occurred in 11 patients: 6 (60%) in the SEPS group, 2 (20%) in the biodegradable group and 3 (30%) in the FCSEMS group (P = 0.16). A total of 8/30 patients (26.6%) were dysphagia-free after the end of follow-up: 1 (10%) in the SEPS group, 3 (30%) in the biodegradable group and 4 (40%) in the FCSEMS group (P = 0.27). More reinterventions were required in the SEPS group (n = 24) than in the biodegradable group (n = 13) or the FCSEMS group (n = 13) (P = 0.24). Multivariate analysis showed that stricture length was significantly associated with higher recurrence rates after temporary stent placement (HR = 1.37; 95% CI = 1.08-1.75; P = 0.011).
Temporary placement of a biodegradable stent or of a FCSEMS in patients with RBES may lead to long-term relief of dysphagia in 30 and 40% of patients, respectively. The use of SEPSs seems least preferable, as they are associated with frequent stent migration, more reinterventions and few cases of long-term improvement. Additionally, longer strictures were associated with a higher risk of recurrence.
- Refractory benign esophageal strictures
- Fully covered self-expanding metal stents
- Biodegradable stents
- Self-expanding plastic stents
- Expandable esophageal stents
Benign esophageal strictures are often caused by esophageal reflux, the ingestion of caustic substances, esophageal surgery, and radiation therapy [1–7]. These patients most commonly present with dysphagia [3, 5]. Treatment of benign esophageal strictures with serial endoscopic dilatation using bougies or balloons has been established as a standard therapy [2, 3, 6], and it is associated with an immediate 80-90% success rate of relieving dysphagia [3, 6]. However, 30-60% of benign strictures will recur during long-term follow-up. Furthermore, in complex strictures that are usually longer (>2 cm), tortuous, angulated or have a severely narrowed diameter, the recurrence rate is considerably higher [3, 5, 7]. In these patients, repeated sessions of dilatation without long-term clinical success are associated with increased the risk of developing complications, patient discomfort and recurrent dysphagia; thus, an alternative treatment strategy should be considered [2, 3, 7]. Surgical procedures, including gastric pull-up and enteral replacement, are potentially curative, but they are associated with high rates of morbidity and mortality, and many patients are not be good surgical candidates or willing to undergo surgery [1, 3, 6]. Theoretically, temporary placement of expandable esophageal stents permits a longer-lasting dilatation effect, maintains luminal patency and simultaneously stretches the stricture [3–6]. Previous studies have proposed the use of temporary self-expandable metal stents (SEMS) with discouraging results due to the high rate of complications, including embedding of the uncovered metal portion of the stent in the esophageal wall, new stricture formation, necrosis and ulcerations from the stents themselves [2–9].
To avoid complications of partially covered/uncovered stents, temporary placement of 3 different types of expandable stents have been used for the treatment of refractory benign esophageal strictures (RBES): self-expanding plastic stents (SEPSs), biodegradable stents and fully covered SEMSs. Several studies have evaluated the clinical effectiveness of SEPSs [2, 4, 6, 10–17]. Although initial studies with SEPSs showed promising results pertaining to dysphagia relief in the great majority of patients [10, 11], recent studies have shown less favorable outcomes, with a clinical success rate well below 50% [2, 4, 6, 12–14]. Furthermore, SEPS are also associated with various complications, especially high stent migration rates [2, 11, 13, 14]. Biodegradable stents have recently been developed and can serve as an alternative for SEPS. Saito et al. reported results from 2 series of patients who received poly-l-lactic esophageal stents [18, 19]. In one study Saito et al. observed a high migration rate (10/13 cases), although no symptoms of re-stenosis were observed within the follow-up period in all cases . Three recent studies have used a novel stent (Ella esophageal stent) composed of the biodegradable polymer polydiaxone [4, 20, 21]. In these 3 studies authors observed low migration rates with the Ella stent (range: 0–22.2%), and encouraging clinical results with clinical success rates ranging from 33% to 60%. One study comparing the Ella stent with the Polyflex stent observed similar clinical performance and complication rates between the 2 stents as well as a significantly lower reintervention rate for the Ella stent group . Finally, an alternative to SEPSs and biodegradable stents in the treatment of RBES is offered by fully covered SEMSs [1, 3, 22, 23]. However, a recent meta-analysis  did not include studies with biodegradable stents, and only one study compared SEPSs and biodegradable stents . Thus, no study has included or compared the 3 types of stents available for the treatment of RBES, and no case series have addressed the outcomes of patients treated with a fully covered Wallflex stent.
This study compared 3 groups of consecutive patients with RBES who underwent temporary placement of SEPSs, biodegradable stents and fully covered SEMSs. Data were collected to analyze the technical success and clinical outcomes of the stents as evaluated by recurrent dysphagia, complications and reinterventions.
Patients and setting
Between July 2005 and March 2011, 3 consecutive cohorts of patients with RBES were enrolled in the study and followed prospectively. One of 3 different types of stents were placed, at different times, in 30 consecutive patients: a SEPS, a biodegradable stent or a fully covered SEMS. The stent used was chosen accordingly with the practice at that time in the participating centre. The placement of the stents was done consecutively (e.g. 10 SEPS, then 10 biodegradable stents and then 10 fully cover SEMS), and not at the discretion of the endoscopist. Patients with esophageal fistulas or leaks, suspicion of malignancy or an upper esophageal sphincter within 3 cm of the stricture were excluded from the study. This study was conducted at a total of 4 referral academic centres (Pulido Valente Hospital, Faculty of Medical Sciences, Lisbon, Portugal; Cuf Infanto Santo Hospital, Lisbon, Portugal; Santo António dos Capuchos Hospital, Lisbon, Portugal; and Garcia de Orta Hospital, Almada, Portugal). All patients provided informed written consent prior to undergoing stent placement. Each institutional review board involved approved this study.
End points and definitions
The primary end point was clinical success defined as the resolution of dysphagia symptoms after the end of long-term follow-up (grade 0–1). Secondary end points included technical success, safe removal of the stents, complications and the need for reinterventions. Dysphagia was graded using a previously described scale as follows: grade 0 = able to eat normal diet/no dysphagia; grade 1 = able to swallow some solid foods; grade 2 = able to swallow only semi solid foods; grade 3 = able to swallow liquids only; grade 4 = total dysphagia . RBES were defined, using a previously defined definition, as the inability to achieve a lumen diameter of 14 mm over five sessions at 2-week intervals or as a result of an inability to maintain a satisfactory luminal diameter for 4 weeks once the target diameter of 14 mm had been achieved . This definition applies only when there is no inflammation at the stricture site. Technical success or successful stent placement was defined as deployment of the stent across the lesion with patency visualized both fluoroscopically and endoscopically. Complications were defined as any adverse event related to stent placement. Complications were also categorized as major (severe events such as perforations, hemorrhage necessitating transfusion, severe pain, or aspiration) and minor (non-life-threatening, such as mild chest pain, globus, reflux symptoms, stent migration, or tissue hyperplasia). Stent migration was defined as either radiographic evidence of the stent within the stomach or endoscopic visualization of the stent having moved from the initial placement location. Reintervention was defined as any procedure performed after initial stent placement, which included stent removal or stent repositioning due to migration, dilatation, additional stent placement and surgery, but did not include scheduled stent removal.
Technique and stents
Before stent placement, all patients underwent an esophagogram. All procedures were performed with patients in the left lateral position under sedation with propofol administered by an anesthesiologist. A guidewire was passed through the stricture, and dilatation was performed as needed using either a balloon dilator or a Savary-Guilliard dilator at the discretion of the operator. The RBES were dilated large enough only to allow passage of the stent delivery apparatus (e.g. 12–14 mm for the SEPS, 9.4 mm for the Ella stent and 6.2 mm for the fully covered SEMS). After dilatation, if needed, internal radiopaque markers were placed at the distal and proximal edges of the stricture. Stents were deployed under fluoroscopic guidance. Endoscopy was repeated immediately after stent placement for the purpose of visualizing the proximal part of the stent to assess adequate placement and deployment of the stent. Resumption of oral intake was permitted on the same day, and patients were cautioned about aspiration risk when necessary. All patients were placed on proton pump inhibitors after stent placement.
The follow-up continued from stent insertion until at least 8 months after stent removal/degradation/migration in all patients. Endoscopies (to detect asymptomatic migration and/or hyperplastic tissue overgrowth), assessments of dysphagia and investigations of any potentially stent-related symptoms were conducted at 1, 2 and 3 months. After stent removal/stent degradation, patients were contacted on a monthly basis and asked to grade their dysphagia using the scale previously described in this manuscript. After 6 months, patients were submitted to endoscopy and then followed on a 2-month basis. In the case of recurrent dysphagia during follow-up, patients contacted their medical assistant in the institution for analysis and reintervention when needed.
The intention-to-treat method was used in all analyses. The χ2 test, Kruskal-Wallis test and the Mann–Whitney U test were used to calculate the statistical significance of different demographic and clinical variables when appropriate. Dysphagia scores, taken at baseline, at 4 weeks and after stent removal/dysphagia recurrence, were compared within each stent group using the Wilcoxon signed-rank test. The dysphagia-free period (esophageal patency) during follow-up after stent removal/degradation/migration was evaluated by the Kaplan-Meier method, and groups were compared using the log-rank test. Multivariate Cox proportional hazard models with forward selection were used to evaluate the multivariate factors potentially affecting the dysphagia-free period after temporary stenting. Age, sex, type of stent, length and location (upper + lower esophagus, middle esophagus or anastomotic) of stricture were the variables included in the analysis. A Poisson regression was conducted to determine possible factors affecting the number of reinterventions. All reported P-values were for two-sided test, and a P-value less than 0.05 was considered to be statistically significant. Statistical analysis was performed using SPSS (Statistical Package for the Social Sciences) 18 (IBM Corporation, New York, USA).
Patient demographics, characteristics of strictures, indications for stent placement and baseline dysphagia scores
SEPS (n = 10)
BD Stent (N = 10)
FCSEMS (n = 10)
Sex, n (%)
Age (years), mean (range)
Characteristics of stricture
Lenght (cm), mean (range)
Location, n (%)
Indication for stent placement,
Dysphagia score before stent placement, mean (±SD)
Stent placement, removal and degradation
Technical success, clinical outcome, dysphagia evolution and reinterventions after temporary placement of 3 different self-expanding stents for the treatment of refractory benign esophageal strictures
SEPS (n = 10)
BD Stent (N = 10)
FCSEMS (n = 10)
Technical success, n (%)
Dysphagia score, mean (±SD)
At 4 weeks
Time to dysphagia recurrence (weeks), mean (range)
Final outcome, n (%)
Follow-up (months), median (range)
Clinical effectiveness and evaluation of dysphagia
Minor and major complications after temporary placement of 3 different self-expanding stents for the treatment of refractory benign esophageal strictures
SEPS (n = 10)
BD Stent (N = 10)
FCSEMS (n = 10)
Total, n in number of patients, n (%)
9 in 7 (70)
7 in 5 (50)
6 in 6 (60)
Minor, n (%)
Moderate chest pain
Tissue hyperplasia (in or overgrowth)
Major, n (%)
Severe chest pain
Follow-up and reinterventions
Overall, 22 patients (73.3%) were submitted to reinterventions (Table 2), which included dilatation, repeat stenting and surgery. More reinterventions were required in the SEPS group (n = 24) than in the biodegradable stent group (n = 13) or the group with temporary placement of a fully covered SEMS (n = 13). These differences were not significant based on the Kruskal-Wallis test (P = 0.24). However, Poisson regression showed that fully covered SEMSs and biodegradable stents resulted in a significantly lower number of reinterventions (B = −0.87; P < 0.05). Migration was observed in 6 patients after SEPS placement. One patient underwent surgery, and the other 5 patients underwent further stenting with either SEPSs (3 patients, 7 stents) or fully covered SEMSs (2 patient, 2 stents). In the 3 patients who were submitted to further stenting with SEPSs, the number 7 stents refers to the total number of stent re-placements; in these patients we did not place a new stent, but we repositioned stents that had migrated. Two patients are still being treated with temporary stent placement (fully covered SEMSs, which migrated again and were repositioned), and 3 patients underwent surgery. Three patients had their SEPSs removed after 3 months and developed recurrent dysphagia. One is still being treated with repeat dilatations, and 2 required surgery. After surgery, one patient had a post-surgical stricture successfully managed by repeated dilatations. In the biodegradable group, stent migration was observed in 2 patients. One patient required surgery, and a post-surgical stricture developed after 3 months. The patient was unsuccessfully managed with a biodegradable stent and is currently under monthly dilatation therapy. The other patient was submitted to repeat dilatations and finally to further stenting with a fully covered SEMS, which migrated into the stomach and was repositioned twice. There were five patients in whom a biodegradable stent completely dissolved in place; they each presented with recurrent dysphagia. Two required surgery and were dysphagia-free (score 0–1), and the remaining 3 are still being treated with repeat dilatations. Stent migration occurred in 3 patients after temporary placement of a fully covered SEMS. Two patients were submitted to further stenting with a fully covered SEMSs (2 stents per patient with new migrations and repositioning of the stent), and the third patient is being treated with repeat dilatations. Of the 3 patients with successful fully covered SEMS removal (one patient had his stent removed at 3 weeks due to globus) and dysphagia recurrence, one patient was further treated with another fully covered stent for 6 months and was dysphagia-free (score 0–1) at 12 weeks after removal of the second stent. One patient was treated with repeat dilatations, and the remaining patient required surgery and is dysphagia-free (score 0–1). For the 22 patients in whom clinical success was not achieved, the mean time to dysphagia recurrence after stent removal/migration/degradation was 4 weeks (range: 2–9 weeks), and there were no significant differences in the time to recurrence in patients treated with SEPSs (mean: 4.3 weeks, range: 2–9 weeks), biodegradable stents (mean: 3.8 weeks, range: 2–9 weeks) or fully covered SEMSs (mean: 3.8 weeks, range: 2–8 weeks) (P = 0.75) [SEMS vs. biodegradable stent (P = 0.95); SEPS vs. fully covered SEMS (P = 0.61); biodegradable stent vs. fully covered SEMS (P = 0.54)].
According to our findings in a prospective, nonrandomized study of 3 consecutive cohorts of patients with RBES, temporary placement of a biodegradable stent or a fully covered SEMS may lead to a long-term dysphagia-free period in 30 and 40% of patients, respectively. The use of SEPS seems to be least preferable, as it is associated with frequent stent migration, more reinterventions and few cases (10%) of long-term improvement. A long stricture was the only significant factor associated with a higher recurrence rate after temporary stent placement.
Unfortunately, unsuccessful management of benign esophageal strictures by serial endoscopic dilatation using bougies or balloons occurs, and the management of these refractory strictures has been considered challenging [4, 5, 7, 27]. Surgery can provide definitive treatment but has been associated with considerable mortality and morbidity, including the development of new anastomotic strictures [2–9]. Partially covered metal stents, which are highly effective in the palliation of malignant esophageal strictures, are associated with high complication rates and are not recommended for RBES . Thus, in a group of patients who have few treatment options available, temporary stent placement can be considered, especially if the procedure is associated with a low complication rate. However, stent placement should be reserved for true refractory strictures, as defined earlier in the Methods section . In our study, all patients were first submitted to serial dilatations during a number of sessions ranging from at least 5 to 10 attempts (some patients were referred from community hospitals where the number of dilatations was greater than 20). Furthermore, all patients were further submitted to several sessions (3 to 6) of intralesional 4-quadrant steroid injections. As proposed by Siersema et al. [4, 5, 7, 27], temporary stent placement was a late step in the treatment strategy of benign esophageal strictures.
Case series evaluating the outcomes of self-expandable plastic stents for benign esophageal diseases
Esophageal strictures (n)
Clinical success n (%)
Evrard et al. 
Migration 9/17 (52.9%)
Repici et al. 
Migration 1/15 (6.7%)
Martin et al. 
Migration 1/18 (5.5%)
Holm et al. 
5/83 interventions (6%)
Migration 53/83 stents placed (63.9%)
Dua et al. 
Migration 8/40 (22%) Perforation 2/40 (5%) Bleeding 3/40 (7.5%) Mortality 1/40 (2.5%)
Oh et al. 
Migration 4/13 (30.8%)
Triester et al. 
Migration 3/5 (60%) Perforation 1/5 (20%) Severe chest pain 1/5 (20%)
Barthel et al. 
Migration 11/13 (85%)
Regarding clinical success of SEPS, the results of our study are clearly in line with those of recent studies [2, 6, 12–14]. At the end of the follow-up period, only 1 patient was dysphagia free. Migration was observed in 60% of our patients. Again, this is in accordance with some studies that have reported migration rates between 52.9 and 85% [2, 11, 13, 14]. One explanation for our high migration rate could be the duration of the study (3 months); longer stenting times are associated with higher risks of migration [2, 11]. In our study, half (n = 3) of the patients had a stent migration after 4 weeks. Alternatively, another explanation could be associated with the result that 8 patients had strictures located in the distal esophagus or in anastomotic areas, as these locations have been associated with higher migration rates for SEPS . Clinically significant tissue hyperplasia was not observed in our study as one might expect from a fully covered device, and this result is in line with the literature [2, 4, 6, 10–16] and also with a recent pooled analysis where tissue in- and overgrowth was uncommon .
Theoretically, a biodegradable stent is ideal for the treatment of RBES as it becomes embedded in the esophageal wall, reducing migration rates, dissolving spontaneously after placement and decreasing the need for reinterventions to remove the stent. Furthermore, a stent made from a material with good tissue compatibility should overcome/reduce the in- and overgrowth of reactive tissue. Three recent studies have reported preliminary results with the new Ella stent [4, 20, 21]. An initial prospective study with 21 patients demonstrated a long-term improvement in 9/21 of the patients (42.9%) . Migration occurred in 2 patients (9.5%), and clinically significant tissue hyperplasia was observed in 1 patient (4.75%). Severe post-stenting pain developed in 3 patients, and minor bleeding was observed in 1 patient. In a prospective study comparing Polyflex stents with ELLA stents, the temporary placement of a biodegradable stent was curative in 6/18 patients (33%) . Major complications occurred in 4 patients (2 had severe retrosternal pain and 2 had hemorrhaging). Clinically significant tissue in- and overgrowth was observed in 2/18 patients (11%). A recent prospective case series of 10 patients who underwent Ella stent placement for benign esophagogastric anastomotic strictures showed a clinical success rate of 60% . No migration was observed, and no complications were observed. Signs of tissue hyperplasia were noticed in 4 patients; however, this was associated with reobstruction and symptoms of dysphagia in only 2 patients.
In our study, 30% of patients in the biodegradable stent group were dysphagia-free at the end of the follow-up period, which is in accordance with 2 previously described studies that have reported a clinical success between 33 and 42.9% [4, 20]. Our migration rate was 20%; however, although this was similar to the rate reported by van Boeckel et al. , this result was unexpected. We anticipated that the embedding of the uncovered stent into the esophagus would prevent migration. In our study, migration occurred in 2 initial patients, and both had long strictures. Eventually, it was found that length selection of the stent was not the best preventative measure of early migration. After the 2 episodes of migration, we began to perform larger balloon dilatation of the stent immediately after deployment to further embed the stent into the esophageal mucosa and reduce the risk of migration. The implementation of balloon dilatation of the Ella stent after deployment did decrease migration rates compared to before balloon dilatation. No other cases of migration were observed and we suggest that this measure can help to mitigate migration after Ella stent deployment. Clinically significant hyperplasia (ingrowth) was observed in 3/10 patients (1 bleeding, 2 reobstructions); this rate was slightly higher than the rates from the 3 studies previously described [4, 20, 21]. The uncovering of the stent may facilitate hyperplasia along with the eventual chemical reaction of the esophageal mucosa with the polydioxanone. However, the low rate of in- and overgrowth found in the literature and in this study suggests that the biodegradable stent has good tissue compatibility.
Removable, fully covered SEMS are an alternative to SEPS and biodegradable stents. Their fully covered design is thought to induce less reactive hyperplasia, making these stents easier to remove and facilitating stenting over longer periods [3, 23, 28]. Few published studies, mostly retrospective, have looked at the outcomes of the use of fully covered SEMSs, and they have reported variable data. One interventional radiology study evaluated the effectiveness of temporary stenting in 55 patients . Three different types of stents were left in place from 1 week to 6 months, and long-term symptom relief was reported in 31% (17/55) of patients. Migration and tissue hyperproliferation were observed in 25 and 31% of patients, respectively. Using the Alimaxx stent, Eloubeidi et al. reported their results from a study of 19 patients . Clinical success was achieved in 21% of patients, no tissue hyperplasia was reported and migration rates were 37%. A recent study retrospectively reviewed 24 patients with refractory post-esophagogastric anastomosis strictures . The stents were removed within 4–8 weeks after placement, and 17/24 patients (70.8%) were free from dysphagia at the end of the follow-up period (12 months) as evaluated by intention-to-treat analysis. Tissue hyperplasia was not observed, and migration was observed in only 1 patient (4.2%). Well-tolerated chest pain and reflux symptoms were the most frequent complications reported.
Our study is the first prospective case series to report the outcomes of treatment with a fully covered Wallflex stent. This stent is easy to implant and can be safely removed. However, complications do occur, and migration was the most common complication observed. Migration is frequent for fully covered stents because of the reduced anchoring capacity of these devices [1, 3, 4, 28]. Migration occurred in 3 patients (30%), which is in accordance with the results of 2 previously cited reports (25 and 37%) [1, 3]. Compared with the SEPS used in our study, which are also fully covered stents, the migration rate of the Wallflex stent was reduced by a factor of 2 (30% vs. 60%). We believe that this result was related to the larger diameters, stronger radial force and the design of the Wallflex stent, which has 2 flared ends. In contrast, the Polyflex stent has only 1 flared end. Another concern with temporary stent placement is tissue hyperplasia. The fact that these stents do not become embedded in the esophageal wall makes clinically significant tissue in- and overgrowth less likely to occur. In our study of 10 patients, no clinically significant tissue hyperplasia was observed, which is in accordance with other reports [3, 23]. Only interventional radiology studies have reported significant tissue hyperplasia [1, 22], which should be interpreted with caution because there was no endoscopic visualization of such tissue hyperproliferation. Furthermore, in our study, one patient was submitted to further stenting over 6 months, and, after that period, the stent was safely removed without signs of hyperplasia. Overall, 4 patients (40%) had long-term resolution of their symptoms after stent removal, which is comparable to the results of 4 earlier studies reporting different clinical successes: 31% , 21% , 48%  and 70.8%  (average: 42.5%).
Although not statistically significant, the results of biodegradable stents or fully covered SEMSs were superior to those of SEPSs in terms of several variables. For example, the duration that patients were dysphagia-free after temporary stenting with a biodegradable stent or a fully covered SEMS was almost 2-fold higher than after placement of a SEPS. While stents were in place, there was a significant difference in the dysphagia scores compared to baseline scores for all stents; all stents were equally effective. However, when comparing baseline dysphagia to post-treatment scores, a significant difference was found in the fully covered group and the biodegradable group but not in the SEPS group; this is a consequence of the differing clinical success rates of the 3 stents. Overall, in the SEPS group, only 1 patient experienced long-term improvement. An explanation for this could be the high rate of SEPS migration when compared with the other 2 stents used. Patients with fully covered or biodegradable stents had their stents in place for longer periods, which may indicate that longer stenting times correlate with higher success rates [4, 10, 11]. The optimal time of stenting is unknown and has been a subject of debate as clear guidelines have yet to emerge. In our multivariate analysis, the length of the stricture was the only factor associated with higher recurrence rates after temporary stent placement. Thus, it is possible that longer strictures would require longer treatment as suggested elsewhere [1, 7, 22, 29]. The optimal duration of temporary stenting can also be influenced by the underlying cause of injury or by the time since the injury to the esophagus occurred [22, 29].
Stenting is safe. In our study, with the exception of migrations, the number of complications was low and, and the complications that occurred were well tolerated. Only 2/30 patients (6%) had significant complications (1 case of bleeding and 1 case of severe pain), which is consistent with most published studies using different expandable stents [2–4, 10–12, 14, 15, 17, 20, 21, 23, 24, 30]. The safety of the procedure in association with reasonable clinical success in a group of patients that is very difficult to manage makes temporary stent placement an appealing option.
Poisson regression showed that fully covered SEMSs and biodegradable stents resulted in a significantly lower number of reinterventions. More reinterventions were required in the SEPS group, which was due to the higher migration rate along with the lower clinical success rate of SEPSs. One of the reinterventions in all groups was restenting. The clinical effectiveness of restenting is limited. While stents are in place, patients benefit from another dysphagia-free period, but after stent removal, the number of patients who experience long-term clinical success is low. This result is consistent with several studies [1, 2, 12, 14, 22, 23]. In our study, restenting was performed in 9 patients, and only one was dysphagia-free after the restenting procedure was performed. In those patients whose primary stent migrated, the second stent generally migrated as well, regardless of whether it was the same or a different fully covered stent, which is in line with other studies in the literature [1, 2, 14, 22]. After migration, restenting with fully covered devices is of poor value; therefore, we suggest restenting with a biodegradable stent which seems to be a better option, as it is associated with a lower migration rate due to the potential embedding of the biodegradable stent in the esophagus.
We suggest that further prospective randomized studies are needed to compare biodegradable stents with fully covered SEMS to determine clinical effectiveness, optimal duration of stenting, value of restenting for longer periods, cost and patient satisfaction.
The present study has several limitations. First, our nonrandomized design could have introduced sampling bias. However, there were no major differences in location, etiology or stricture length between the 3 groups studied. Another potential weakness is the different time to follow-up for the 3 groups. The SEPS group had a longer follow-up period, which could have influenced the final outcome because, theoretically, the longer the follow-up period the higher the probability of dysphagia recurrence. However, in our study, all patients had at least 8 months of follow-up after stent removal/degradation/migration, and most other studies showed that the benefits of temporary stenting decreased rapidly with time for non-responders [1, 12, 20, 21, 23]. In our study, no patient had recurrent dysphagia after 9 weeks following stent removal/migration/degradation (Figure 3). Thus, different follow-up times were not likely to be associated with the poor results found in the SEPS group. Our study had a relatively low number of patients per group, which limited the study’s statistical strength. However, most of the studies using self-expanding stents have small population sizes due to the low incidence of RBES, and studies with a patient population large enough to have adequate power to detect minor outcome differences are generally impractical. Finally, this study was performed in tertiary referral centers, and we cannot exclude the possibility of obtaining a poorer outcome in community hospitals.
The strengths of our study are the prospective design with a relatively long-term follow-up. To our knowledge, this is the first study to compare the 3 types of self-expanding stents available for the treatment of RBES and the first case series to report the outcomes of patients treated with a fully covered Wallflex stent.
In conclusion, our data suggest that the temporary placements of biodegradable stents and fully covered SEMSs have similar utility in the treatment of RBES, as they were associated with clinical success in 1/3 of our patients. Of the 3 available stents, SEPSs were associated with the worst clinical success rate as well as with a higher number of migrations and reinterventions. Taken together, temporary stenting for RBES may be useful, especially in those patients in whom other therapeutic options are unavailable. However, better stents and different strategies are needed to overcome the difficulties of managing these patients, particularly when a long stricture is present.
- Kim JH, Song HY, Choi EK, Kim KR, Shin JH, Lim JO: Temporary metallic stent placement in the treatment of refractory benign esophageal strictures: results and factors associated with outcome in 55 patients. Eur Radiol. 2009, 19: 384-390. 10.1007/s00330-008-1151-2.View ArticlePubMedGoogle Scholar
- Holm AN, Levy JG, Goustout CJ, Topazian MD, Baron TH: Self-expanding plastic stents in treatment of benign esophageal conditions. Gastrointest Endosc. 2008, 67: 20-25. 10.1016/j.gie.2007.04.031.View ArticlePubMedGoogle Scholar
- Eloubeidi MA, Talreja JP, Lopes TL, Al-Awabdy BS, Shami VM, Kahaleh M: Success and complications associated with placement of fully covered removable self-expandable metal stents for benign esophageal diseases (with videos). Gastrointest Endosc. 2011, 73: 673-681. 10.1016/j.gie.2010.11.014.View ArticlePubMedGoogle Scholar
- van Boeckel PG, Vleggaar FP, Siersema PD: A comparison of temporary self-expanding plastic and biodegradable stents for refractory benign esophageal strictures. Clin Gastroenterol Hepatol. 2011, 9: 653-659. 10.1016/j.cgh.2011.04.006.View ArticlePubMedGoogle Scholar
- Siersema PD: Stenting for benign esophageal strictures. Endoscopy. 2009, 41: 363-373. 10.1055/s-0029-1214532.View ArticlePubMedGoogle Scholar
- Dua KS, Vleggaar FP, Santharam R, Siersema PD: Removable self-expanding plastic esophageal stent as a continuous, non-permanent dilator in treating refractory benign esophageal strictures: a prospective two-center study. Am J Gastroenterol. 2008, 103: 2988-2994. 10.1111/j.1572-0241.2008.02177.x.View ArticlePubMedGoogle Scholar
- Siersema PD, Wijkerslooth LR: Dilatation of refractory benign esophageal strictures. Gastrointest Endosc. 2009, 70: 1000-1012. 10.1016/j.gie.2009.07.004.View ArticlePubMedGoogle Scholar
- Ackroyd R, Watson DI, Devitt PG, Jamieson GG: Expandable metallic stents should not be used in the treatment of benign esophageal strictures. J Gastroenterol Hepatol. 2001, 16: 484-487. 10.1046/j.1440-1746.2001.02367.x.View ArticlePubMedGoogle Scholar
- Sanda GS, Marcon NE: Expandable metal stents for benign esopagheal obstruction. Gastrointest Endosc Clin N Am. 1999, 9: 437-446.Google Scholar
- Repici A, Conio M, De Angelis C, Battaglia E, Musso A, Pellicano R, Goss M, Venezia G, Rizzetto M, Saracco G: Temporary placement of an expandable polyester silicone-covered stent for treatment of refractory benign esophageal strictures. Gastrointest Endosc. 2004, 60: 513-519. 10.1016/S0016-5107(04)01882-6.View ArticlePubMedGoogle Scholar
- Evrard S, Le Moine O, Lazaraki G, Dormann A, El Nakadi I, Deviere J: Self-expanding plastic stents for benign esophageal lesions. Gastrointest Endosc. 2004, 60: 894-900. 10.1016/S0016-5107(04)02278-3.View ArticlePubMedGoogle Scholar
- Oh YS, Kochman ML, Ahmad NA, Ginsberg GG: Clinical outcomes after self-expanding plastic stent placement for benign strictures. Dig Dis Sci. 2010, 55: 1344-1348. 10.1007/s10620-010-1134-4.View ArticlePubMedGoogle Scholar
- Triester SL, Fleischer DE, Sharma VK: Failure of self-expanding plastic stents in treatment of refractory benign esophageal strictures. Endoscopy. 2006, 38: 533-537. 10.1055/s-2006-925318.View ArticlePubMedGoogle Scholar
- Barthel JS, Kelley ST, Klapman JB: Management of persistent gastroesophageal anastomotic strictures with removable self-expandable polyester silicon-covered (Polyflex) stents: an alternative to serial dilatation. Gastrointest Endosc. 2008, 67: 546-552. 10.1016/j.gie.2007.10.047.View ArticlePubMedGoogle Scholar
- Martin RC, Woodall C, Duvall R, Scoggins CR: The use of self-expanding silicone stents in esophagectomy strictures: less cost and more efficiency. Ann Thorac Surg. 2008, 86: 436-440. 10.1016/j.athoracsur.2008.04.039.View ArticlePubMedGoogle Scholar
- Garcia-Cano J: Dilatation of benign strictures in the esophagus and colon with the polyflex stent: a case series study. Dig Dis Sci. 2008, 53: 341-346. 10.1007/s10620-007-9864-7.View ArticlePubMedGoogle Scholar
- Repici A, Hassan C, Sharma P, Conio M, Siersema P: Systematic review: the role of self-expanding plastic stents for benign esophageal strictures. Alim Pharm Ther. 2010, 31: 1268-1275. 10.1111/j.1365-2036.2010.04301.x.View ArticleGoogle Scholar
- Saito Y, Tanaka T, Andoh A, Minematsu H, Hata K, Tsujikawa T, Nitta N, Murata K, Fujiyama Y: Novel biodegradable stents for benign esophageal strictures following endoscopic submucosal dissection. Dig Dis Sci. 2008, 53: 330-333. 10.1007/s10620-007-9873-6.View ArticlePubMedGoogle Scholar
- Saito Y, Tanaka T, Andoh A, Minematsu H, Hata K, Tsujikawa T, Nitta N, Murata K, Fujiyama Y: Usefulness of biodegradable stents constructed of poly-l-lactic acid monofilaments in patients with benign esophageal stenosis. World J Gastroenterol. 2007, 13: 3977-3980.View ArticlePubMedPubMed CentralGoogle Scholar
- Repici A, Vleggaar FP, Hassan C, van Boeckel PG, Romeo F, Pagano N, Malesci A, Siersema PD: Efficacy and safety of biodegradable stents for refractory benign esophageal strictures: the BEST (Biodegradable Esophageal Stent) study. Gastrointest Endosc. 2010, 72: 927-934. 10.1016/j.gie.2010.07.031.View ArticlePubMedGoogle Scholar
- van Hooft JE, Henegouwen MI, Rauws EA, Bergman JJ, Busch OR, Fockens P: Endoscopic treatment of benign anastomotic esophagogastric strictures with a biodegradable stent. Gastrointest Endosc. 2011, 73: 1043-1047. 10.1016/j.gie.2011.01.001.View ArticlePubMedGoogle Scholar
- Song HY, Jung HY, Park SI, Kim SB, Lee DH, Kang SG, Il Min Y: Covered retrievable expandable nitinol stents in patients with benign esophageal strictures: Initial experience. Radiology. 2000, 217: 551-557.View ArticlePubMedGoogle Scholar
- Liu J, Hu Y, Cui C, Li Y, Lin X, Fu J: Removable, fully covered, self-expandable metal stents for the treatment of refractory benign esophagogastric anastomotic strictures. Dysphagia. 2012, 27: 260-264. 10.1007/s00455-011-9361-1.View ArticlePubMedGoogle Scholar
- Thomas T, Abrams KR, Subramanian V, Mannath J, Ragunath K: Esophageal stents for benign refractory strictures: a meta-analysis. Endoscopy. 2011, 43: 386-393. 10.1055/s-0030-1256331.View ArticlePubMedGoogle Scholar
- Knyrim K, Wagner H, Bethge N, Keymling M, Vakil N: A controlled trial of an expansile metal stent for palliation of esophageal obstruction due to inoperable cancer. N Engl J Med. 1993, 329: 1302-1307. 10.1056/NEJM199310283291803.View ArticlePubMedGoogle Scholar
- Kockman ML, McClave SA, Boyce HW: The refractory and the recurrent esophageal stricture: a definition [Letter to Editor]. Gastrointest Endosc. 2005, 62: 474-475.View ArticleGoogle Scholar
- Wijkerslooth LR, Vleggaar FP, Siersema PD: Endoscopic management of difficult or recurrent esophageal strictures. Am J Gastroenterol. 2011, 106: 2080-2091. 10.1038/ajg.2011.348.View ArticlePubMedGoogle Scholar
- Sharma P, Kozarek R: Role of esophageal stents in benign and malignant diseases. Am J Gastroenterol. 2010, 105: 258-273. 10.1038/ajg.2009.684.View ArticlePubMedGoogle Scholar
- Siersema PD, Hirdes MM: What is the optimal duration of stent placement for refractory, benign esophageal strictures?. Nat Clin Pract Gastroenterol Hepatol. 2009, 6: 146-147.View ArticlePubMedGoogle Scholar
- Wong RF, Adler DG, Hilden K, Fang JC: Retrievable esophageal stents for benign indications. Dig Dis Sci. 2008, 53: 322-329. 10.1007/s10620-007-9883-4.View ArticlePubMedGoogle Scholar
- The pre-publication history for this paper can be accessed here:http://www.biomedcentral.com/1471-230X/12/70/prepub
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.