- Research article
- Open Access
- Open Peer Review
DPO multiplex PCR as an alternative to culture and susceptibility testing to detect Helicobacter pylori and its resistance to clarithromycin
© Lehours et al; licensee BioMed Central Ltd. 2011
- Received: 26 July 2011
- Accepted: 17 October 2011
- Published: 17 October 2011
Macrolide resistance in Helicobacter pylori is the major risk factor for treatment failure when using a proton pump inhibitor-clarithromycin containing therapy. Macrolide resistance is due to a few mutations on the 23S ribomosal subunit encoded by the 23S rRNA gene. The present study aimed at investigating the performance of the dual priming oligonucleotide (DPO)-PCR kit named Seeplex® ClaR-H. pylori ACE detection designed to detect H. pylori and two types of point mutations causing clarithromycin resistance in H. pylori.
The performance of Seeplex® ClaR-H. pylori ACE detection was evaluated on 127 gastric biopsies in comparison to conventional bacterial culture followed by the determination of susceptibility to clarithromycin by E-test, as well as by an in-house real-time PCR using a fluorescence resonance energy transfer (FRET) technology.
Considering culture as the reference test, the sensitivity of DPO-PCR and real-time FRET-PCR was 97.7% and 100% while specificity was 83.1% and 80.7%, respectively. However, both PCR were concordant in detecting 14 H. pylori positive cases which were negative by culture. Globally, E-test and DPO-PCR were concordant with regard to clarithromycin susceptibility in 95.3% of the cases (41/43), while real-time FRET-PCR and DPO-PCR were concordant in 95% (57/60).
The DPO-PCR is an interesting tool to detect H. pylori on gastric biopsies and to study its susceptibility to clarithromycin in laboratories that cannot perform real-time PCR assays.
- Fluorescence Resonance Energy Transfer
- Gastric Biopsy
- Macrolide Resistance
- Clarithromycin Resistance
Macrolide resistance in Helicobacter pylori is the major risk factor for treatment failure when using a proton pump inhibitor (PPI)-clarithromycin containing therapy . Macrolide resistance is due to a few mutations on the 23S ribomosal subunit encoded by the 23S rRNA gene [2, 3]. These mutations (A2142C, A2142G, A2143G), are easy to detect by numerous molecular methods directly on gastric biopsy specimens and even on stool samples [4–7].
A new PCR format named DPO-PCR for "Dual Priming Oligonucleotide" was recently developed . DPO-PCR is a multiplex PCR assay that increases specificity and sensitivity of detection compared to conventional PCR, by blocking non-specific binding sites therefore eliminating imperfect primer annealing. This new technology can be used for many applications in the field of in vitro diagnostics: simultaneous detection of multiple pathogens and of polymorphisms (SNPs), as well as simultaneous genotyping of multiple pathogen subtypes. DPO-PCR is based on a multiplex PCR using a DPO patented technology . The structure of the DPO primers is fundamentally different from that of conventional primers. Indeed, the primer is divided into two parts by a 5 polydeoxyinosine linker which allows a more specific hybridization at temperatures between 55 and 65°C. This linker forms a "bubble-like structure" which itself is not involved in priming, rather it delineates the boundary between two parts. It therefore generates two recognition reactions of the primer on the target sequence. According to the manufacturer (see http://www.seegene.com/en/research/core_020.php), the 5' end (approximately 20 bases) binds preferentially to the matrix and initiates stable annealing acting as a "stabilizer". The 3' end is shorter (approximately 10 bases) and binds afterwards to the target site but only if the first step has taken place without a mismatch. The 3' end determines a target-specific extension and acts as a "determiner". Therefore, although the longer 5'-segment binds to a non-target site, the shorter segment resists non-specific extension. The short 3'-portion alone fails to make a priming at an annealing temperature. The latter also binds preferentially to the target and avoids non-specific binding. This PCR can be performed in any conventional thermocycler.
The performance of this PCR format for the detection of H. pylori 23S rDNA mutations, involved in macrolide resistance was previously evaluated in a study published in 2007 by Woo et al.,  with a 94.1% concordance between the DPO-based multiplex PCR and culture followed by a phenotypic susceptibility test.
In an article by Cho AR and Lee MK in Korean language, they also compared this method to culture and histology, and concluded that it could be used for the diagnosis of H. pylori infection and the determination of clarithromycin resistance . However, they used a disk diffusion method which is not a generally accepted technique for testing H. pylori antimicrobial susceptibility. The present study is a retrospective study performed by the National Reference Centre for Helicobacters in France which aimed at investigating the performance of the Seeplex® ClaR-H. pylori ACE detection kit (Seegene, Seoul, Korea) in comparison to standard phenotypic tests as well as the real-time fluorescence resonance energy transfer (FRET)-PCR developed and routinely used in our laboratory .
The Seeplex® ClaR-H. pylori ACE detection kit was evaluated retrospectively on DNAs extracted from 127 gastric biopsies received at the French National Reference Centre for Helicobacters (Bordeaux, France) during the year 2009. There was no preselection according to the gastric site. Consecutive biopsies were included until about half of the number of biopsies positive for H. pylori was attained.
The performance of the kit was compared to conventional bacterial culture followed by the determination of susceptibility to clarithromycin by E-test, and an in-house real-time PCR detection using the FRET technology .
H. pylori strains were obtained from the corresponding gastric biopsies, after culture on Wilkins-Chalgren agar plates (Oxoid, Dardilly, France) supplemented with human blood (10% v/v) and antibiotics (10 μg/ml of vancomycin, 10 μg/ml of cefsulodin, 5 μg/ml of trimethoprim, and 10 μg/ml of amphotericin B) under microaerobic conditions, as already described [11, 12]. Forty-four culture positive cases were finally included.
Phenotypic susceptibility testing
Susceptibility to clarithromycin was assessed using the E-test method (bioMérieux, Marcy l'Etoile, France) performed as previously described  and using the EUCAST breakpoints: S ≤ 0.25 μg/ml; R > 0.5 μg/ml (http://www.eucast.org/clinical_breakpoints/).
DNA extraction from gastric biopsies
Genomic DNA from gastric biopsies was extracted by using the MagnaPure LC DNA Isolation Kit I and the MagnaPure LC Isolation Station (Roche Applied Science, Penzberg, Germany). DNAs were stored at -20°C until required for analysis.
The real-time FRET-PCR is designed to detect clarithromycin susceptible H. pylori (wild type) and the mutations responsible for clarithromycin resistance: A4142G and A2143G, without distinguishing between them, as well as A2142C. This test was performed as previously described .
DPO-PCR was performed using the Seeplex® ClaR-H. pylori ACE detection kit according to the manufacturer's recommendations (Seegene distributed by Eurobio Laboratoires, Courtaboeuf, France) and analyzed using a semi-automated system called Screen tape® allowing an ultra rapid migration and analysis of the PCR products in small polyacrylamide gels. 8-methoxysporalen was added during the mix preparation to intercalate between double-stranded nucleic acids generated during amplification, thereby limiting carry-over contamination after UV irradiation and before PCR product analysis. The Seeplex® ClaR-H. pylori ACE detection kit includes 3 primer pairs with a DPO structure which allows amplification of the H. pylori 23S rDNA (621 bp amplicon) and detection of A2142G and A2143G mutations (194 bp and 475 bp, respectively). The kit also includes a primer pair for internal control.
DPO-PCR is a multiplex PCR that can be performed in any standard thermocycler.
Briefly, histological preparations were stained with hematoxylin and eosin and Giemsa stains and the presence of H. pylori was evaluated according to the Sydney system. Histological results were used only in case of discrepant results obtained between DPO-PCR and FRET-PCR.
Evaluation of Sensitivity and Specificity
The proportion of positives by DPO-PCR among the true positives defined the sensitivity and the proportion of negatives by DPO-PCR among the true negatives defined the specificity.
Concerning the 127 biopsies included in the study, culture was positive for H. pylori in 44 cases (34.6%), the real-time FRET-PCR in 60 cases (47.2%), and the DPO-PCR in 57 cases (44.9%). H. pylori status obtained from histological diagnosis was available for only 89 patients.
Global results obtained for different diagnosis tests for the detection of Helicobacter pylori in human gastric biopsies
Culture (n = 44)
FRET-PCR (n = 60)
DPO-PCR (n = 57)
Total (n = 127)
Using culture as the reference test, the sensitivity of DPO-PCR and real-time FRET PCR was 97.7% and 100%, respectively, and the specificity was 83.1% and 80.7%, respectively.
Comparison of Helicobacter pylori susceptibility to clarithromycin by E-test, real-time FRET-PCR and DPO-PCR on human gastric biopsies
Total (n = 60)
For the 26 macrolide resistant isolates, a 23S rDNA mutation was detected in 25 biopsies by both PCR formats. One biopsy was considered as a wild type by real-time FRET-PCR but was negative by DPO-PCR (histology was also negative).
Globally, E-test and DPO-PCR were concordant in 95.3% of these cases (41/43).
We found a good correlation for the detection of H. pylori and the detection of clarithromycin susceptibility between the DPO-PCR and the real-time FRET-PCR routinely used in our Reference Centre. Overall, the performance is very good for a non-real-time PCR format. Compared to other PCR formats developed to detect mutations involved in macrolide resistance for H. pylori, DPO-PCR requires no investment in additional technical or expensive detection devices. One disadvantage is that users must run the detection of PCR fragments themselves on a 2% agarose gel before analyzing the PCR bands obtained, compared to real-time PCR formats available to date where PCR amplification is monitored automatically. DPO-PCR is therefore more time-consuming. However, in our study the semi-automated system called ScreenTape® was used. ScreenTape® simplifies the analysis of the results of this multiplex PCR assay.
The cost of the test is highly dependent on the activity and equipment of the laboratory in which the test is performed; however, it is significantly higher than the cost of the in-house method tested in parallel in the present study.
DPO-PCR detected more H. pylori positive biopsies than culture alone, with an excellent correlation with the FRET-PCR. Woo et al., identified 49 H. pylori positive samples among 165 culture-negative specimens using DPO-PCR . This result leads us to believe that there is no specificity problem regarding DPO-PCR, rather a problem of sensitivity regarding culture.
The excellent correlation between DPO-PCR and E-test susceptibility is in line with the previous study published by Woo et al., where they found a 94.1% concordance between both methods . In the work of Woo et al., two strains categorized as susceptible by E-test appeared resistant by DPO-PCR. Cho et al., also described that the results of PCR and E-test on 3 of the 8 mutation-positive biopsies were discrepant . In the present study, DPO-PCR detected resistances missed by E-test also in two cases (Table 2). This could be explained by the detection limit of the Seeplex® ClaR-H. pylori ACE detection kit which is 100 copies/reaction (100 copies/3 μl DNA). According to Woo HY et al., DPO-PCR can detect mutants present among wild-type strains at a level as low as 2% and more than 100 copies/20 μl . For such a low proportion, the E-test method missed a resistant strain.
As indicated in the Materials and Methods, the Seeplex® ClaR-H. pylori ACE detection kit includes 3 primer pairs with a DPO structure which allows amplification of the H. pylori 23S rDNA (621 bp amplicon) and detection of the A2142G and A2143G mutations (194 bp and 475 bp, respectively). The first primer pair is designed to hybridize regardless of the presence of any mutation inside the PCR fragments. In the case of the A2142G mutation, its specific primer hybridizes and generates a 194 bp PCR product with the reverse H. pylori 23S rDNA primer. In the case of the A2143G mutation, its specific primer hybridizes and forms a 475 bp PCR product with the forward H. pylori 23S rDNA primer. Therefore, it is not possible to distinguish between 1) gastric biopsies containing a mixture of a wild-type strain and a mutated strain and 2) biopsies containing only a mutated strain (the 621 bp band corresponding to amplification of H. pylori 23S rDNA is almost always present). It has no practical consequences because the detection of a resistant population is sufficient to exclude macrolides from the eradication therapy to be implemented.
Moreover, for 22 biopsies where A2143G was detected by DPO-PCR, the corresponding 475 bp amplicon was alone in only 4 cases which means that for these corresponding DNAs the reverse H. pylori 23S rDNA apparently failed to hybridize and to generate the additional 621 bp amplicon. We believe that this is the reason why, in some rare cases, false negatives by DPO-PCR can occur (a total of 3 in the present study). The primer pairs which allow the amplification of the H. pylori 23S rDNA could be slightly modified to avoid this problem.
Users should keep in mind that whenever possible H. pylori culture should be performed, and only in cases where standard microbiology fails, the use of molecular methods are really indicated. The rationale behind this is that not only clarithromycin resistance is of interest but also that of other antimicrobials like tetracycline, quinolones, rifamycins and metronidazole. However, the Seeplex® ClaR-H. pylori ACE detection kit is an excellent molecular test to detect H. pylori in gastric biopsies and to study its sensitivity to clarithromycin, especially in laboratories without expertise in culturing this bacterium and without a real-time PCR apparatus. At a time when clarithromycin resistance is increasing (prevalence is >20% in many countries), clinical laboratories could be enticed by this new PCR format.
The authors thank Leila Labadi and Salha Ben Amor (Université de Bordeaux, Centre National de Référence des Campylobacters et des Hélicobacters, Bordeaux, France) for technical assistance.
- Malfertheiner P, Megraud F, O'Morain C, Bazzoli F, El-Omar E, Graham D, Hunt R, Rokkas T, Vakil N, Kuipers EJ: Current concepts in the management of Helicobacter pylori infection: the Maastricht III Consensus Report. Gut. 2007, 56 (6): 772-781. 10.1136/gut.2006.101634.View ArticlePubMedGoogle Scholar
- Occhialini A, Urdaci M, Doucet-Populaire F, Bebear CM, Lamouliatte H, Megraud F: Macrolide resistance in Helicobacter pylori: rapid detection of point mutations and assays of macrolide binding to ribosomes. Antimicrob Agents Chemother. 1997, 41 (12): 2724-2728.PubMedPubMed CentralGoogle Scholar
- Menard A, Santos A, Megraud F, Oleastro M: PCR-restriction fragment length polymorphism can also detect point mutation A2142C in the 23S rRNA gene, associated with Helicobacter pylori resistance to clarithromycin. Antimicrob Agents Chemother. 2002, 46 (4): 1156-1157. 10.1128/AAC.46.4.1156-1157.2002.View ArticlePubMedPubMed CentralGoogle Scholar
- Agudo S, Alarcon T, Urruzuno P, Martinez MJ, Lopez-Brea M: Detection of Helicobacter pylori and clarithromycin resistance in gastric biopsies of pediatric patients by using a commercially available real-time polymerase chain reaction after NucliSens semiautomated DNA extraction. Diagn Microbiol Infect Dis. 2010, 67 (3): 213-219. 10.1016/j.diagmicrobio.2010.02.021.View ArticlePubMedGoogle Scholar
- Burucoa C, Garnier M, Silvain C, Fauchere JL: Quadruplex real-time PCR assay using allele-specific scorpion primers for detection of mutations conferring clarithromycin resistance to Helicobacter pylori. J Clin Microbiol. 2008, 46 (7): 2320-2326. 10.1128/JCM.02352-07.View ArticlePubMedPubMed CentralGoogle Scholar
- Cambau E, Allerheiligen V, Coulon C, Corbel C, Lascols C, Deforges L, Soussy CJ, Delchier JC, Megraud F: Evaluation of a new test, genotype HelicoDR, for molecular detection of antibiotic resistance in Helicobacter pylori. J Clin Microbiol. 2009, 47 (11): 3600-3607. 10.1128/JCM.00744-09.View ArticlePubMedPubMed CentralGoogle Scholar
- Oleastro M, Menard A, Santos A, Lamouliatte H, Monteiro L, Barthelemy P, Megraud F: Real-time PCR assay for rapid and accurate detection of point mutations conferring resistance to clarithromycin in Helicobacter pylori. J Clin Microbiol. 2003, 41 (1): 397-402. 10.1128/JCM.41.1.397-402.2003.View ArticlePubMedPubMed CentralGoogle Scholar
- Chun JY, Kim KJ, Hwang IT, Kim YJ, Lee DH, Lee IK, Kim JK: Dual priming oligonucleotide system for the multiplex detection of respiratory viruses and SNP genotyping of CYP2C19 gene. Nucleic Acids Res. 2007, 35 (6): e40-10.1093/nar/gkm051.View ArticlePubMedPubMed CentralGoogle Scholar
- Woo HY, Park DI, Park H, Kim MK, Kim DH, Kim IS, Kim YJ: Dual-priming oligonucleotide-based multiplex PCR for the detection of Helicobacter pylori and determination of clarithromycin resistance with gastric biopsy specimens. Helicobacter. 2009, 14 (1): 22-28. 10.1111/j.1523-5378.2009.00654.x.View ArticlePubMedGoogle Scholar
- Cho AR, Lee MK: [A comparison analysis on the diagnosis of Helicobacter pylori infection and the detection of clarithromycin resistance according to biopsy sites]. Korean J Lab Med. 2010, 30 (4): 381-387. 10.3343/kjlm.2010.30.4.381.View ArticlePubMedGoogle Scholar
- Megraud F: A growing demand for Helicobacter pylori culture in the near future?. Ital J Gastroenterol Hepatol. 1997, 29 (6): 574-576.PubMedGoogle Scholar
- Megraud F, Lehours P: Helicobacter pylori detection and antimicrobial susceptibility testing. Clin Microbiol Rev. 2007, 20 (2): 280-322. 10.1128/CMR.00033-06.View ArticlePubMedPubMed CentralGoogle Scholar
- The pre-publication history for this paper can be accessed here:http://www.biomedcentral.com/1471-230X/11/112/prepub
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.