Back-to-Back Comparison of Auto-Fluorescence Imaging (AFI) Versus High Resolution White Light Colonoscopy for Adenoma Detection
- Kentaro Moriichi1,
- Mikihiro Fujiya1Email author,
- Ryu Sato2, 3,
- Jiro Watari3, 4,
- Yoshiki Nomura1,
- Toshie Nata1,
- Nobuhiro Ueno1,
- Shigeaki Maeda3,
- Shin Kashima1, 3,
- Kentaro Itabashi1,
- Chisato Ishikawa1,
- Yuhei Inaba1,
- Takahiro Ito1,
- Kotaro Okamoto1,
- Hiroki Tanabe1,
- Yusuke Mizukami1,
- Yusuke Saitoh5 and
- Yutaka Kohgo1
© Moriichi et al.: licensee BioMed Central Ltd. 2012
Received: 12 October 2011
Accepted: 22 June 2012
Published: 22 June 2012
Some patients under close colonoscopic surveillance still develop colorectal cancer, thus suggesting the overlook of colorectal adenoma by endoscopists. AFI detects colorectal adenoma as a clear magenta, therefore the efficacy of AFI is expected to improve the detection ability of colorectal adenoma. The aim of this study is to determine the efficacy of AFI in detecting colorectal adenoma.
This study enrolled 88 patients who underwent colonoscopy at Asahikawa Medical University and Kushiro Medical Association Hospital. A randomly selected colonoscopist first observed the sigmoid colon and rectum with conventional high resolution endosopy (HRE). Then the colonoscopist changed the mode to AFI and handed to the scope to another colonoscopist who knew no information about the HRE. Then the second colonoscopist observed the sigmoid colon and rectum. Each colonoscopist separately recorded the findings. The detection rate, miss rate and procedural time were assessed in prospective manner.
The detection rate of flat and depressed adenoma, but not elevated adenoma, by AFI is significantly higher than that by HRE. In less-experienced endoscopists, AFI dramatically increased the detection rate (30.3%) and reduced miss rate (0%) of colorectal adenoma in comparison to those of HRE (7.7%, 50.0%), but not for experienced endoscopists. The procedural time of HRE was significantly shorter than that of AFI.
AFI increased the detection rate and reduced the miss rate of flat and depressed adenomas. These advantages of AFI were limited to less-experienced endoscopists because experienced endoscopists exhibited a substantially high detection rate for colorectal adenoma with HRE.
KeywordsAutofluorescence imaging Colorectal adenoma Detection rate Flat and depressed adenoma Less-experienced endoscopist High-resolution colonoscope
Colorectal cancer is one of the most common malignant tumors in Eastern and Western countries . During the process of colon carcinogenesis, normal epithelia are thought to initially turn into benign adenomas, accumulate gene alterations and then transform to advanced adenocarcinomas . All adenoma are considered to be premalignant lesions. The elimination of colon adenoma is therefore an effective strategy to prevent the development of colon cancer. Several trials on an endoscopic resection for colon adenoma successfully decreased the mortality of colon cancer . However, some patients under close colonoscopic surveillance still develop colorectal cancer . This discrepancy may be caused by the rapid progression of adenomas as well as the overlooking of colorectal adenoma. Indeed, systematic reviews of back-to back colonoscopies showed that 15% to 32% of colorectal adenomas were possibly missed by colonoscopy , particularly flat and depressed adenomas . Advanced endoscopic instruments may therefore decrease the miss rate of adenomas and optimize the potential for colorectal cancer prevention.
AFI is a novel technology which can capture fluorescence (500–630 nm) emitted from intestinal or other tissues. This device produces an excitation light source of 442 nm, delivers it to the tissue surface and then captures the reflection and fluorescence by two high-sensitivity CCDs. Those captured signals are respectively transformed into red or blue colors, and then are displayed on the monitor as a color image in real-time . Fluorescence imaging is thought to be an efficient tool for the evaluation of dysplasia in Barrett esophagus , esophageal and gastric cancer  and dysplasia , as well as for making a differential diagnosis of intestinal lymphoma  and also for assessing the inflammation activity in ulcerative colitis . Three studies proposed the controversial results concerning the usefulness of AFI on the detection of colon lesions. Matsuda et al. proposed the predominance of AFI for detecting polyps of the proximal colon compared to white light endoscopy using a modified back-to-back method . In contrast, the other two studies showed that AFI did not reduce the adenoma miss-rate using two inspections of the entire colon by conventional high-resolution endoscopy (HRE) or AFI . There seems to be some bias with regard to these results, since the endoscopic findings of HRE and AFI could not be assessed independently because each colonoscopic examination was performed by a single endoscopist in those studies. Therefore, the efficacy of AFI in detecting colorectal adenoma still remains to be elucidated.
The current study aimed to assess the efficacy of AFI for improving the colorectal adenoma detection rate by comparing experienced and less experienced endoscopists in a prospective manner.
Demographics of the patients and detected lesions
M : F
45 : 43
Mean age (y.o.)
64.0 ± 10.0
Indication for colonoscopy
Indication for colonoscopy (symptoms: screening: surveillance)
28 : 38 : 22
Number of lesions
5.8 ± 6.5
Type 0-IIb or 0-IIc
3.7 ± 1.8
Flat & depressed
Colonoscopic procedure and randomization
All patients were prepared using 2 L of polyethylene glycol solution in this study. None of them exhibited insufficient bowel preparation. All procedures were performed by one of either 5 experienced endoscopists (> 8000 standard and > 100 AFI) or 3 less-experienced endoscopists (< 500 standard and < 10 AFI). Because this study aimed to elucidate the differences in the significance of AFI between experienced and less-experienced endoscopists, we selected highly experienced endoscopists and endoscopists with much less experience. A colonoscopist who was randomly selected with a sealed envelope method first observed the sigmoid colon and rectum (for 40 cm from anal verge) with HRE. Next, the colonoscopist changed the mode for AFI and handed to the scope to another colonoscopist who was not aware of the information of HRE. The second colonoscopist who was also selected using a sealed envelope method inserted a colonoscope for 40 cm and observed the sigmoid colon and rectum under sufficient filling with air. Each colonoscopist separately recorded the colonoscopic findings including the location, color and shape of the detected lesions (Figure 1). Thereafter, both colonoscopists cooperated to insert the scope again and obtain biopsy specimens from all detected lesions. The biopsied specimens were histologically diagnosed by a pathologist with no clinical and endoscopic information according to the Vienna classification . The entire procedural time for each procedure was separately measured and recorded for all the patients. The detected lesions were classified into two groups, elevated type (0-IIa) and flat or depressed type (0-IIb or -IIc) according to Paris classification . The detection rate and miss rate of colorectal adenoma by all participants, experienced or less-experienced endoscopists were assessed. The ratio of the cases with adenoma divided by the all cases enrolled was defined as the detection rate. The miss rate was defined as the ratio of all detected adenoma divided by the adenoma as diagnosed by each procedure. The primary goal of this study was to determine the detection ability of AFI in comparison to that of HRE. The second goal was to explore whether the usefulness of AFI for detecting colon lesions was influenced by the morphological features of colon lesions and the experience of the endoscopist. These two secondary outcomes were both exploratory.
To determine the sample size, our preliminary data suggested that the miss rates of HRE and AFI for the detection of colon adenoma were 25.0% and 4.2%, respectively. When the alpha error was defined as 0.05 (two-sided) and the power as 0.8, the number of cases required was estimated to be 66 (33 for each group). Twenty-five percent of the enrolled cases were thought to drop out of the study, so the target number of cases was 85. A total of 88 patients were enrolled and analyzed in this study.
The Wilcoxon signed-rank test was applied for the statistical analysis of the detection and miss rates, and for the procedural time in all participants, because the data were paired and non-parametric. The Mann–Whitney U-test was used to perform the statistical analysis of such data by either experienced or less-experienced endoscopists, because the data were unpaired and non-parametric. A p value < 0.05 was considered to be statistically significant. For the purposes of the statistical analysis, individual lesions were assumed to constitute statistically independent observations even when more than one lesion was present in a single patient.
The detection rate of colorectal adenoma w5th HRE and AFI
Cases with adenoma detected
Detection rates (%)
p < 0.05
Flat and depressed adenoma
p < 0.05
Cases examined by experienced endoscopists
Cases examined by less experienced endoscopists
p < 0.05
HRE was performed by experienced endoscopists in 54 of 88 cases and by less-experienced endoscopists in 34 cases. AFI was performed by experienced endoscopists in 55 cases and by less-experienced endoscopists in 33 cases. In 24 of 88 cases, one or more adenomas, which were histologically classified into category 3 or 4.1 according to the Vienna classification , were detected by either HRE or AFI. No lesions corresponding to false positive regions, such as normal mucosa or vascular abnormalities, were detected by either HRE or AFI. Whereas HRE detected one or more colorectal adenomas in 16 of 88 cases (18.2%), AFI detected colorectal adenoma in 23 cases (26.1%). The detection rate for AFI is therefore significantly higher than that for HRE (p < 0.05). AFI detected 8 0-IIb or 0-IIc adenomas in 88 cases while conventional colonoscopy detected only 3 (p < 0.05). In contrast, no difference was observed between HRE and AFI regarding the detection rate of 0-IIa adenomas.
The miss rate of colorectal adenoma with HRE and AFI
Number of lesions
Miss rates (%)
p < 0.0001
p < 0.05
p < 0.0001
p < 0.0001
p < 0.0001
Less experienced endoscopists
p < 0.0001
p < 0.05
p < 0.005
Twenty-nine adenomas were detected by either HRE or AFI in 88 cases. Twenty-one of the 29 adenomas detected from the 88 patients were captured by HRE and 28 were captured by AFI. The miss rate of HRE (27.6%) was significantly higher than that of AFI (3.4%; p < 0.05). The miss rate of HRE (50.0%) was much higher than that of AFI (0%) in less-experienced endoscopists while no significant deference in the miss rate was seen between HRE (24%) and AFI (5.3%) in experienced endoscopists, thus suggesting the efficacy of AFI in detecting colorectal adenoma by less-experienced endoscopists.
The current study demonstrated that AFI improved the detection rate of colorectal adenoma, particularly flat and depressed adenomas, but not for elevated adenomas. An elevated adenoma is easily detected as a reddish and protruding lesion by HRE because HRE can detects abnormalities through capturing the changes of color and shape. However, flat and depressed adenoma is sometimes described just a faint red areas with no elevation, which thus tend to be difficult to detect by HRE. On the other hand, AFI captures the fluorescence mostly emitted from collagen in the submucosal layer and diagnoses the abnormal area through a diminished fluorescence intensity. The dysplastic grade of the lesions is the main factor that affects the fluorescence captured by AFI . The increase in dysplasia appears to facilitate the detection rate of colorectal adenoma by AFI, regardless the shape of the lesion.
A noteworthy finding of the current study is that the efficacy of AFI to detect colorectal adenoma was limited for less-experienced endoscopists. The experienced endoscopists appeared to empirically recognize many of adenomas, even flat and depressed adenoma, with HRE, and thus AFI was not a superior disgnostic modality for experienced endoscopists in detecting colorectal adenoma. The present study revealed that the miss rate with HRE by experienced endoscopists (24.0%) is relatively lower than that previously reported . This was probably because only highly experienced endoscopists (> 8000 standard and > 100 AFI) were strictly selected as experienced endoscopists in this study. Conversely, it should not be easy for less-experienced endoscopists to find colorectal adenoma (50.0%), particularly flat and depressed adenoma, with HRE. Most endoscopists, including less-experienced endoscopists, can immediately utilize AFI because AFI diagnosis is simply based on the color intensity of magenta. This objectivity of AFI probably facilitated the detection rate and reduced the miss rate of colorectal adenoma in less-experienced endoscopists. The clinical value of HRE and AFI is therefore considered to depend on the experience of endoscopists.
Three previous reports have shown controversial results concerning the efficacy of AFI for detecting colorectal adenoma. Matsuda et al. concluded that AFI showed a higher detection rate than did white light endoscopy when using a modified back to back method wherein the authors observed the proximal colon twice, once with white light endoscopy and once with AFI, with both examinations being performed by the same endoscopist . Subsequently, two investigations proposed the detection rate of AFI and HRE using the back to back method. These studies inspected the entire colon twice during withdrawal: once with AFI and once with HRE by the same endoscopist and found no efficiency of AFI for reducing the adenoma miss-rate in comparison to HRE . These studies possess a potential bias because both endoscopic examinations are performed by the same endoscopist and the second endoscopic diagnosis was therefore influenced by the findings of the first endoscopy. This may explain the controversial results obtained from the two similar studies. Conversely, in the current study, AFI was performed by an endoscopist who was not aware of the information of HRE. This allowed the determination of the efficiency of AFI in the detection of flat and depressed adenoma by less-experienced endoscopists.
Chromoendoscopy can improve the detection of colonic lesions [31–36]. However, chromoendoscopy with magnified observation is time-consuming, operator-dependent and impossible to switch back to the conventional colonoscopy, which may affect the ability to detect mucosal abnormalities in other areas. Narrow band imaging (NBI), a new system in which spectral features are modified by narrowing the bandwidth of spectral transmittance with optical filters and can assess capillary architecture and microvessels just by manipulating a button , is also a promising tool to improve the detection rate of colorectal adenoma . Indeed, recent studies have demonstrated adenoma miss-rates for HRE of 40% to 46% when a second examination was performed with NBI. This suggests that NBI is superior to HRE when utilized for adenoma detection . Because NBI has a high ability to discriminate colon adenoma from hyperplasia , the combination of AFI and NBI can potentially improve the selective detection rate of colon adenoma. Further analysis to compare the efficacy of AFI, NBI or their combination for detecting colon neoplasms is therefore needed to establish the optimal diagnostic modality to perform screening colonoscopy using new advanced imaging systems such as NBI and AFI.
The procedural time to perform AFI (267.0 ± 20.8 seconds) was significantly longer than that of HRE (144.5 ± 8.8 seconds). This is a disadvantage of AFI when using it as screening examinations. Whereas AFI can capture the fluorescence emitted from intestinal tissue, the resolution of AFI is still not sufficient. Therefore, endoscopists have to be cautious when performing colonoscopic examinations with AFI, in order not to increase the overall examination time. New advances in endoscopic tools are expected to produce a higher resolution and optimized operation system of AFI, thus leading to a reduction in the time needed to perform the AFI procedures.
Because it is not easy to judge the lesions detected by both HRE and AFI, or either of the two procedures (particularly when multiple lesions are detected), when the entire colon is targeted, this study investigated only the recto-sigmoid colon. This is thought to be a limitation of this study. A further analysis covering the entire colon will be warranted to draw conclusions about the overall usefulness of AFI for the detection of colon lesions.
The current study demonstrated that AFI increased the detection rate and reduced the miss rate of colorectal adenoma, particularly flat and depressed adenomas. However, this advantage of AFI was limited to the less-experienced endoscopists because experienced endoscopists exhibited a substantially low miss rate of colorectal adenoma with HRE. Further advances in endoscopic instruments will provide an AFI with a high resolution and optimized operation system, thereby preventing the lengthy procedural time which is a disadvantage of AFI.
K.M. and M.F. provided major input into the conceptual development of the studies, wrote the manuscript and supervised all investigations. R.S., J.W., Y.N., T.N., N.U., S.K., K.I., Y.I. and H.T. performed the endoscopic examinations. Y.I. assessed the histological findings. S.M., C.I., T.I. and K.O. managed and treated the enrolled patients and collected and analyzed the data. Y.M., Y.S. and Y.K. helped design studies, interpret the data, and prepare/review the manuscript. All authors read and approved the final manuscript.
- Potter JD: Colorectal cancer: molecules and populations. J Natl Cancer Inst. 1999, 91: 916-932. 10.1093/jnci/91.11.916.View ArticlePubMed
- Fearon ER, Vogelstein B: A genetic model for colorectal tumorigenesis. Cell. 1990, 61: 759-767. 10.1016/0092-8674(90)90186-I.View ArticlePubMed
- Lengauer C, Kinzler K, Vogelstein B: Genetic instabilities in human cancers. Nature. 1998, 396: 643-649. 10.1038/25292.View ArticlePubMed
- Winawer SJ, Zauber AG, Ho MN, O'Brien MJ, Gottlieb LS, Sternberg SS, Waye JD, Schapiro M, Bond JH, Panish JF, Ackroyd F, Shike M, Kurtz RC, Hornsby-Lewis L, Gerdes H, Stewart ET, National Polyp Study Workgroup: Prevention of Colorectal Cancer by Colonoscopic Polypectomy. N Eng J Med. 1993, 329: 1977-1981. 10.1056/NEJM199312303292701.View Article
- Robertson DJ, Greenberg ER, Beach M, Sandler RS, Ahnen D, Haile RW, Burke CA, Snover DC, Bresalier RS, McKeown-Eyssen G, Mandel JS, Bond JH, Van Stolk RU, Summers RW, Rothstein R, Church TR, Cole BF, Byers T, Mott L, Baron JA: Colorectal cancer in patients under close colonoscopic surveillance. Gastroenterology. 2005, 129: 34-41. 10.1053/j.gastro.2005.05.012.View ArticlePubMed
- van Rijn JC, Reitsma JB, Stoker J, Bossuyt PM, van Deventer SJ, Dekker E: Polyp miss rate determined by tandem colonoscopy: a systematic review. Am J Gastroenterol. 2006, 101: 343-350. 10.1111/j.1572-0241.2006.00390.x.View ArticlePubMed
- Fujii T, Rembacken BJ, Dixon MF, Yoshida S, Axon AT: Flat adenomas in the United Kingdom: are treatable cancers being missed?. Endoscopy. 1998, 30: 437-443. 10.1055/s-2007-1001304.View ArticlePubMed
- Rembacken BJ, Fujii T, Cairns A, Dixon MF, Yoshida S, Chalmers DM, Axon AT: Flat and depressed colonic neoplasms: a prospective study of 1000 colonoscopies in the UK. Lancet. 2000, 355: 1211-1214. 10.1016/S0140-6736(00)02086-9.View ArticlePubMed
- Saitoh Y, Waxman I, West AB, Popnikolov NK, Gatalica Z, Watari J, Obara T, Kohgo Y, Pasricha PJ: Prevalence and distinctive biologic features of flat colorectal adenomas in a North American population. Gastroenterology. 2001, 120: 1657-1665. 10.1053/gast.2001.24886.View ArticlePubMed
- Tsuda S, Veress B, Tóth E, Fork FT: Flat and depressed colorectal tumours in a southern Swedish population: a prospective chromoendoscopic and histopathological study. Gut. 2002, 51: 550-555. 10.1136/gut.51.4.550.PubMed CentralView ArticlePubMed
- Takehana S, Kaneko M, Mizuno H: Endoscopic Diagnostic System Using Autofluorescence. Diagnostic and Therapeutic Endoscopy. 1999, 5: 59-63. 10.1155/DTE.5.59.PubMed CentralView ArticlePubMed
- Namihisa A, Miwa H, Watanabe H, Kobayashi O, Ogihara T, Sato N: A new technique: light-induced fluorescence endoscopy in combination with pharmacoendoscopy. Gastrointest Endosc. 2001, 53: 343-348.View ArticlePubMed
- Georgakoudi I, Jacobson BC, Van Dam J, Backman V, Wallace MB, Müller MG, Zhang Q, Badizadegan K, Sun D, Thomas GA, Perelman LT, Feld MS: Fluorescence, reflectance, and light-scattering spectroscopy for evaluating dysplasia in patients with Barrett’s esophagus. Gastroenerology. 2001, 120: 1620-1629. 10.1053/gast.2001.24842.View Article
- Borovicka J, Fischer J, Neuweiler J, Netzer P, Gschossmann J, Ehmann T, Bauerfeind P, Dorta G, Zürcher U, Binek J, Meyenberger C: Autofluorescnce endoscopy in surveillance of Barrett’s esophagus. A multicenter randomized trial on diagnostic efficacy. Endoscopy. 2006, 38: 867-872. 10.1055/s-2006-944726.View ArticlePubMed
- Kara MA, Bergman JJGH: Autofluorescence imaging and narrow-band imaging for the detection of early neoplasia inpatients with Barrett’s esophagus. Endoscopy. 2006, 38: 627-631. 10.1055/s-2006-925385.View ArticlePubMed
- Uedo N, Iishi H, Tatsuta M, Yamada T, Ogiyama H, Imanaka K, Sugimoto N, Higashino K, Ishihara R, Narahara H, Ishiguro S: A novel videoendoscopy system by using autofluorescence and reflectance imaging for diagnosis of esophagogastric cancers. Gastrointest Endosc. 2005, 62: 521-528. 10.1016/j.gie.2005.06.031.View ArticlePubMed
- Messmann H, Endlicher E, Freunek G, Rümmele P, Schölmerich J, Knüchel R: Fluorescence endoscopy for the detection of lowand high grade dysplasia in ulcerative colitis using systemic or local 5-aminolaevulinic acid sensitization. Gut. 2003, 52: 1003-1007. 10.1136/gut.52.7.1003.PubMed CentralView ArticlePubMed
- Matsumoto T, Moriyama T, Yao T, Mibu R, Iida M: Autofluorescence imaging colonoscopy for the diagnosis of dysplasia in ulcerative colitis. Inflamm Bowel Dis. 2007, 13: 640-641. 10.1002/ibd.20104.View ArticlePubMed
- van den Broek FJ, Fockens P, van Eeden S, Reitsma JB, Hardwick JC, Stokkers PC, Dekker E: Endoscopic tri-modal imaging for surveillance in ulcerative colitis: randomised comparison of high-resolution endoscopy and autofluorescence imaging for neoplasia detection; and evaluation of narrow-band imaging for classification of lesions. Gut. 2008, 57: 1083-1089. 10.1136/gut.2007.144097.PubMed CentralView ArticlePubMed
- Ueno N, Fujiya M, Moriichi K, Ikuta K, Nata T, Konno Y, Ishikawa C, Inaba Y, Ito T, Sato R, Okamoto K, Tanabe H, Maemoto A, Sato K, Watari J, Ashida T, Saitoh Y, Kohgo Y: Endosopic Autofluorescence Imaging is Useful for the Differential Diagnosis of Intestinal Lymphomas Resembling Lymphoid Heyperplasia. J Clin Gastroenterol. 2011, 45: 507-513. 10.1097/MCG.0b013e3181fbe22a.View ArticlePubMed
- Fujiya M, Saitoh Y, Watari J, Moriichi K, Kohgo Y: Auto-Fluorescence Imaging is useful to assess the activity of ulcerative colitis. Dig Endosc. 2007, 19: 145-149. 10.1111/j.1443-1661.2007.00725.x.View Article
- Matsuda T, Saito Y, Fu KI, Uraoka T, Kobayashi N, Nakajima T, Ikehara H, Mashimo Y, Shimoda T, Murakami Y, Parra-Blanco A, Fujimori T, Saito D: Does autofluorescence imaging videoendoscopy system improve the colonoscopic polyp detection rate?—a pilot study. Am J Gastroenterol. 2008, 103: 1926-1932. 10.1111/j.1572-0241.2008.01931.x.View ArticlePubMed
- van den Broek FJ, Fockens P, Van Eeden S, Kara MA, Hardwick JC, Reitsma JB, Dekker E: Clinical evaluation of endoscopic trimodal imaging for the detection and differentiation of colonic polyps. Clin Gastroenterol Hepatol. 2009, 7: 288-295. 10.1016/j.cgh.2008.10.025.View ArticlePubMed
- Rotondano G, Bianco MA, Sansone S, Prisco A, Meucci C, Garofano ML, Cipolletta L: Trimodal endoscopic imaging for the detection and differentiation of colorectal adenomas: a prospective single-centre clinical evaluation. Int J Colorectal Dis. 2012, 27: 331-336. 10.1007/s00384-011-1312-7.View ArticlePubMed
- Participants in the Paris Workshop. The Paris endoscopic classification of superficial neoplastic lesions: esophagus, stomach, and colon:. 2003, 58: 3-43.
- Schlemper RJ, Riddell RH, Kato Y, Borchard F, Cooper HS, Dawsey SM, Dixon MF, Fenoglio-Preiser CM, Fléjou JF, Geboes K, Hattori T, Hirota T, Itabashi M, Iwafuchi M, Iwashita A, Kim YI, Kirchner T, Klimpfinger M, Koike M, Lauwers GY, Lewin KJ, Oberhuber G, Offner F, Price AB, Rubio CA, Shimizu M, Shimoda T, Sipponen P, Solcia E, Stolte M, Watanabe H, Yamabe H: The Vienna classification of gastrointestinal epithelial neoplasia. Gut. 2000, 47: 251-255. 10.1136/gut.47.2.251.PubMed CentralView ArticlePubMed
- Moriichi K, Fujiya M, Sato R, Nata T, Nomura Y, Ueno N, Ishikawa C, Inaba Y, Ito T, Okamoto K, Tanabe H, Mizukami Y, Watari J, Saitoh Y, Kohgo Y: Autofluorescence imaging and the quantitative intensity of fluorescence for evaluating the dysplastic grade of colonic neoplasms. Int J Colorectal Dis. 2011, 27: 325-330.View ArticlePubMed
- Rastogi A, Bansal A, Wani S, Callahan P, McGregor DH, Cherian R, Sharma P: Narrow-band imaging colonoscopy—a pilot feasibility study for the detection of polyps and correlation of surface patterns with polyp histologic diagnosis. Gastrointest Endosc. 2008, 67: 280-286. 10.1016/j.gie.2007.07.036.View ArticlePubMed
- East JE, Suzuki N, Stavrinidis M, Guenther T, Thomas HJ, Saunders BP: Narrow band imaging for colonoscopic surveillance in hereditary non-polyposis colorectal cancer. Gut. 2008, 57: 65-70.View ArticlePubMed
- Heresbach D, Barrioz T, Lapalus MG, Coumaros D, Bauret P, Potier P, Sautereau D, Boustière C, Grimaud JC, Barthélémy C, Sée J, Serraj I, D'Halluin PN, Branger B, Ponchon T: Miss rate for colorectal neoplastic polyps: a prospective multicenter study of back-to-back video colonoscopies. Endoscopy. 2008, 40: 284-290. 10.1055/s-2007-995618.View ArticlePubMed
- Brooker JC, Saunders BP, Shah SG, Thapar CJ, Thomas HJ, Atkin WS, Cardwell CR: Williams CB; Total colonic dye-spray increases the detection of diminutive adenomas during routine colonoscopy: a randomized controlled trial. Gastrointest Endosc. 2002, 56: 333-338. 10.1016/S0016-5107(02)70034-5.View ArticlePubMed
- Hurlstone DP, Cross SS, Slater R, Sanders DS, Brown S: Detecting diminutive colorectal lesions at colonoscopy: a randomised controlled trial of pan-colonic versus targeted chromoscopy. Gut. 2004, 53: 376-380. 10.1136/gut.2003.029868.PubMed CentralView ArticlePubMed
- Lapalus MG, Helbert T, Napoleon B, Rey JF, Houcke P: Ponchon T; Société Française d'Endoscopie Digestive: Does chromoendoscopy with structure enhancement improve the colonoscopic adenoma detection rate?. Endoscopy. 2006, 38: 444-448. 10.1055/s-2006-925265.View ArticlePubMed
- Le Rhun M, Coron E, Parlier D, Nguyen JM, Canard JM, Alamdari A, Sautereau D, Chaussade S, Galmiche JP: High resolution colonoscopy with chromoscopy versus standard colonoscopy for the detection of colonic neoplasia: a randomized study. Clin Gastroenterol Hepatol. 2006, 4: 349-354. 10.1016/j.cgh.2005.12.009.View ArticlePubMed
- Kato S, Fujii T, Koba I, Sano Y, Fu KI, Parra-Blanco A, Tajiri H, Yoshida S, Rembacken B: Assessment of colorectal lesions using magnifying colonoscopy and mucosal dye spraying: can significant lesions be distinguished?. Endoscopy. 2001, 33: 306-310. 10.1055/s-2001-13700.View ArticlePubMed
- Fu KI, Sano Y, Kato S, Fujii T, Nagashima F, Yoshino T, Okuno T, Yoshida S, Fujimori T: Chromoendoscopy using indigo carmine dye spraying with magnifying observation is the most reliable method for differential diagnosis between non-neoplastic and neoplastic colorectal lesions: a prospective study. Endoscopy. 2004, 36: 1089-1093. 10.1055/s-2004-826039.View ArticlePubMed
- Gono K, Yamazaki K, Doguchi N, Nonami T, Obi T, Yamaguchi M, Ohyama N, Machida H, Sano Y, Yoshida S, Hamamoto Y, Endo T: Endoscopic observation of tissue by narrow band illumination. Opt Rev. 2003, 10: 211-215. 10.1007/s10043-003-0211-8.View Article
- Gono K, Obi T, Yamaguchi M, Ohyama N, Machida H, Sano Y, Yoshida S, Hamamoto Y, Endo T: Appearance of enhanced tissue features in narrow-band endoscopic imaging. J Biomed Opt. 2004, 9: 568-577. 10.1117/1.1695563.View ArticlePubMed
- Inoue T, Murano M, Murano N, Kuramoto T, Kawakami K, Abe Y, Morita E, Toshina K, Hoshiro H, Egashira Y, Umegaki E, Higuchi K: Comparative study of conventional colonoscopy and pan-colonic narrow-band imaging system in the detection of neoplastic colonic polyps: a randomized, controlled trial. J Gastroenterol. 2008, 43: 45-50. 10.1007/s00535-007-2125-x.View ArticlePubMed
- van den Broek FJ, van Soest EJ, Naber AH, van Oijen AH, Mallant-Hent RCh, Böhmer CJ, Scholten P, Stokkers PC, Marsman WA, Mathus-Vliegen EM, Curvers WL, Bergman JJ, van Eeden S, Hardwick JC, Fockens P, Reitsma JB, Dekker E: Combining autofluorescence imaging and narrow-band imaging for the differentiation of adenomas from non-neoplastic colonic polyps among experienced and non-experienced endoscopists. Am J Gastroenterol. 2009, 104: 1498-1507. 10.1038/ajg.2009.161.View ArticlePubMed
- Sato R, Fujiya M, Watari J, Ueno N, Moriichi K, Kashima S, Maeda S, Ando K, Kawabata H, Sugiyama R, Nomura Y, Nata T, Itabashi K, Inaba Y, Okamoto K, Mizukami Y, Saitoh Y, Kohgo Y: The diagnostic accuracy of high-resolution endoscopy, autofluorescence imaging and narrow-band imaging for differentially diagnosing colon adenoma. Endoscopy. 2011, 43: 862-868. 10.1055/s-0030-1256510.View ArticlePubMed
- The pre-publication history for this paper can be accessed here:http://www.biomedcentral.com/1471-230X/12/75/prepub
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.